The Boat Anchor Manual Archive Path: /hp/a/. This is the BAMA archive. These manuals are available for download and free of charge. I am now accepting. View and Download HP A operating and service manual online. DIGITAL VOLTMETER. A Measuring Instruments pdf manual download. Hp A Pdf User Manuals. View online or download Hp A Operating And Service Manual.
|Published (Last):||25 October 2011|
|PDF File Size:||6.26 Mb|
|ePub File Size:||19.48 Mb|
|Price:||Free* [*Free Regsitration Required]|
Before connecting ac power to the 34S5A.
The DVM may or gp not be within its specification and the only way to tell for sure is to use a more accurate standard. Example C is similar to A with the ex- ception that connecting guard in this manner allows any common-mode current generated between the source low and powerline ground to flow in the measurement circuit.
The uprange points, the downrange points, and the accuracy of the instrument should be kept 34455a mind when making a measurement.
Instrumont Mtaauroffloiit Timoi Remots Control. After the measurement is taken, the 34SSA can be triggered again for a new reading. The resultant measurements arc stored by the main controller to correct measurements made on he.
HP A Digital Voltmeter Operaiting & SERVICE Manual | eBay
These are then connected via flying cables with the rest of the circuit. Auto Range See all condition definitions – opens in a new window or tab Adjust the meter reference controls for a zero reading on the meter of the Test Oscillator and use the level con- trols to maintain this reading whenever the frequency is varied.
The square root amplifier reverses the action of squaring amplifier. The large exponent is he key. Refer to Figure The logic check consists of a Figure illustrates the circuit configuration Measurement.
| HP B – Calibration module from HP A
When the 3455a is the External Trigger mode, the user can trigger the instrument from an external trigger pulse. Simplified Input Switching Diagram With a ten-to-one error reduction, however, these regions are relatively narrow. General Description An equation describing this computation is: It is recommended that the Transfer Standard be calibrated and adjusted just prior to use.
Calibration Cycle See the following example: Turn-Over Errors Schematic 1, 5, and 6 8-G The frequency response of the average converter is from 30 Hz to Hz in the normal ac volts function and from Hz to kHz in the fast ac volts func- tion. Auto Calibration-DC Voltage For rated measurement accuracy, the SA should be allowed to warm up for at least one hour.
Test Procedure Longwood, Florida, United States. The A reading should be within the test limits listed in Tableverifying its volt tenth scale accuracy with High Resolution on. Adjust the meter reference controls for a zero reading on the meter of the Test Oscillator and use the level controls to maintain this reading whenever the frequency is varied.
Ohms Servicing 8-D H sure to wire rear panel guard connections yourself, if- using the rear panel input terminals. During the cur- rent mode of operation the converter supplies a constant current of. Note the 34SSA reading. The A reading should be greater than 0.