Before you When calculating rates.g. Quick Start-Up Guide Setting up an infusion for the Ivac Rate Calculation ml to be infused in 2 hours 30 minutes. The IVAC® Volumetric Pump automatically regulates the infusion rate of intravenous solutions. Users are advised to read, to understand this manual and to be thoroughly familiar with the Pump prior to Tel: (27) (0) interferon α-2α, , , , , , , interferon α-2b, , , , , IVAC, ixabepilone, –,, ,, , ixazomib, .
|Published (Last):||13 August 2008|
|PDF File Size:||16.11 Mb|
|ePub File Size:||17.39 Mb|
|Price:||Free* [*Free Regsitration Required]|
The change of state initiates a countdown to the near end of infusion alarm point in software. For product questions or needle-free manhal educational materials, contact your CareFusion representative. Remove the two screws securing the mains inlet to the lower case. Press plunger plate button. Use the following guide to eliminate the cause of the alarm.
The user is responsible for verifying the suitability of a product used, if it is not recommended by CareFusion. The flow sensor will also be able to detect empty containers. The countdown depends upon the syringe type and rate of infusion. Ibac the next section for further details. Note the position of the joint see the illustration below.
CareFusion IVAC 598 Directions For Use Manual
Check cover detect reed switch operation by running Self-Test During operation, Display goes blank into sleep mode after 2 minutes. For standard nursecall, fit the 2-way nursecall connector into the unmarked PCB connector.
Display uvac Backlight Display test Backlight Check that the display dots are an even tone as they all go on full backlight, then to dim, then turn off. In addition, two flexible printed circuits are utilised to hold the optical sensors and to provide the necessary interconnects to the moving parts of the pump.
Desolder battery from the Manuall PCB. Motor Gearbox Breakdown Note: The display will read EOI opto: See ‘Occlusion Test Force Ranges’ on the following page.
F it and fasten the case halves together and check the EOI point again. Volume mode between 0. The position at which the display changes EOI point depends on the type of syringe in use. On the File menu, select Properties.
Follow the instructions uvac with the individual infusion set. Sale Lots Listing Search for See ‘Configuration Options ‘ on the following pages for further details. Allows protocols to be modified in SET mode.
If the pump alarms or the power is switched off the relay contacts close, activating the nurse call circuit. Hinge the base plate open and remove it. Lay the capacitor horizontally and support by applying a small quantity of silicon sealant. The LED in the optical switch is switched on and off by TR9 so that the operation of the optical device can be self tested during normal operation.
Disconnect de-solder the flexi circuits from each other see Figure 2. View in English Change to English. Check battery terminal connections are secure.
If the user of the pump requires continued operation during power mains interruptions, it is recommended that the pump be powered from manuxl uninterruptible power supply or a battery.
To print the manual completely, please, download it. On earlier pump models, the isolated Nursecall circuit PL3 is used to allow connection to a conventional nurse call system.
Lots in Sale Number | Hilditch Group
Remove the Communications PCB, pulling the ribbon cable through the slot in the case wall. The optical encoder signals are all fed via schmitt trigger inverters in IC Replacing the molex header Procedure: End of Infusion Optical Switch A slotted optical switch is mounted on the transmission motor plate to detect the mqnual end of infusion point. R Remove spacer buttons as required.
DVM Resistance Meter 1. Check motor and mechanism, replace as necessary. A suitable printer must be connected to the pump.